PT100 with amp board
Yes, the additional resolution would be nice.
Hopefully the parts should arrive soon, then I can play with it, I'll hook it up to my Arduino MEGA first to see what ADC values I get, then mod the amp and see what I get again, then test with a K type thermocouple hooked up with the PT100 sensor to compare values and adjust the thermistor (Thermocouple) table for the PRUSA.
The rest of my re-build is going bit by bit, new thicker cables from new PSU to reverse mounted Einsy, also adjusted the new PSU to 26 volts output, I'm going with the PIN10 P3 connector analog in for the PT100 amp.
Designing new cover plate with FAN casing for the LRS-350 PSU with a customisable finger guard to suit fitting an Arctic F9 Silent 92mm fan to the PSU.
The idea is so that the fan housing can be printed separately to the rest of the cover, this way anyone can make their own design to suit their taste, and maybe a skin for the rest of the PSU so that the colour matches with whatever else you want, again customisable designs.
And I'm still learning about the firmware, I may have to break it down into chunks to load onto the MEGA to fully learn what different parts do, but this is a long term project, even crossed my mind that this may be a good way to tune the trinamics, just buy a trinamics module, run barebones software on the MEGA, plug it into one of the steppers on the printer and drive it, should be easier to connect a scope and no danger of frying your EINSY.
should be easier to connect a scope and no danger of frying your EINSY.
?? You should be able to ground at the PS common and not do any damage. At least my first pass at looking at the EINSY schematics show it should be safe - all of the GND points float, so should just drop to earth when attaching the scope. Or are you wanting to look at the stepper drive differentially?
The hot end parts have not arrived yet, so nothing happening yet ☹️
My parts arrived, and I have glued a 10k multiturn potentiometer to the board ready to solder, so that I can adjust the gain from 10 to 16. Potentiometer pins will be soldered directly to the ends of R5.
Currently there is a glass block on it pressing down until the clear gorilla glue dries, and tomorrow I will solder the ends.
All soldered and set to 3.300K Ohms. I had to buy a new multimeter when I realised my existing 30+ years old Fluke one was no longer accurate, tried new battery, but no good, guess I can't complain really.
Got a Beha Amprobeso should last a good number of years, pointless getting a cheap Chinese meter, it would have been worse than my faulty 30+ years old meter. 🙂
Just creating a little snap shut case for the board and testing on an Arduino MEGA can start.